This learning package has been developed for health professionals to develop skills and knowledge and ultimately demonstrate competency in safely setting up, commencing and providing necessary documentation for the NIKI T34TM syringe pump.
The package is relevant to all clinical settings – community, residential aged care and inpatient.
Important Information: NIKI T34TM syringe pump (3rd edition only)
The internal battery that runs the internal real-time clock depletes if the NIKI T34TM (3rd edition only) is not used for several weeks.
If this occurs, when the NIKI T34TM
is next used, the following could occur:
- The internal battery may be completely depleted, and the NIKI T34TM may display a 'SYSTEM ERROR' message on start-up.
- Resolve this issue by restarting the pump. Once the pump has restarted, a 'SEND FOR SERVICE' message will appear. Press . This will clear the 'SEND FOR SERVICE' message and you can use the pump as normal.
- Please read the Technical Bulletin (175kb pdf) for further information.